201316th International Congress of Metrology
|Nombre de pages||5|
|Section||Métrologie électrique / Electrical metrology|
|Publié en ligne||7 octobre 2013|
Traçabilité des mesures des paramètres S aux fréquences intermediaries (100 kHz – 100 MHz)
Laboratoire National de Métrologie et d’Essais, 29 avenue Roger Hennequin, 78197 Trappes, France
a Corresponding author: email@example.com
At LNE, a new 50 Ω primary standard with a female type N connector (7 mm) was just developed for providing one-port traceable S-parameters measurements in this frequency range . The typical expanded uncertainties (k = 2) are in the range 1.10-3 to 4.10-3 and in the range 2,5.10-3 to 1.10-2 for reflection coefficient values close to 0 and above 1.10-2 respectively. To complete the LNE measurement capabilities, a 50 Ω load with a male type N connector is currently being developed which will be establish the traceability of two-ports S-parameters measurements. The development of the 50 Ω load with a male type N connector is presented in this paper.
© Owned by the authors, published by EDP Sciences, 2013
This is an Open Access article distributed under the terms of the Creative Commons Attribution License 2.0, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.