Traçabilite des réseaux d’impedance de ligne (RSIL)
1 LNE, 29 avenue Roger Hennequin 78197 Trappes Cedex, France
2 SIQ, Tržaška c. 2 SI- 1000 Ljubljana, Slovénie
a Email de correspondance de l’auteur : firstname.lastname@example.org
The Line Impedance Stabilization Network (LISN) is mostly used in electromagnetic compatibility for conducted and radiated radio-frequency emission and susceptibility tests. The input impedance is one of the most important parameters in LISN calibration. To measure the input impedance, an adapter is required to connect the Vector Network Analyzer (VNA) to the Equipment Under Test (EUT) port of the LISN. For test laboratories, it is of best interest to perform this calibration with high accuracy. Until now the impact of the adapter has not been clearly demonstrated theoretically. We present the design of an adapter and its associated modelling, a method for removing its impact from the measurement, the theoretical error on magnitude and phase input impedance.
© Owned by the authors, published by EDP Sciences, 2015
This is an Open Access article distributed under the terms of the Creative Commons Attribution License 4.0, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.