201718th International Congress of Metrology
|Number of page(s)||5|
|Section||Dynamics of pressure / Les dynamiques de la pression|
|Published online||18 September 2017|
Force Calibration with Build Up Systems
Hottinger Baldwin Messtechnik GmbH, Im Tiefen See 45 ; 64293 Darmstadt, Germany
The idea behind a build-up system is to use force transducer with a lower capacity in a mechanical parallel arrangement to increase the capacity of the complete system. Three or more individual sensors are completed to a build-up system. The aim of those systems is to calculate the behavior of the complete system from the calibration of the individual sensors.
Key words: Build up systems / force reference transducer / parallel circuit of strain gauge based sensors
© The Authors, published by EDP Sciences, 2017
This is an Open Access article distributed under the terms of the Creative Commons Attribution License 4.0, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.